Back to Home
built RF sampler for the Bird Model 43 Thruline Wattmeter
By Kevin K. Custer W3KKC
$35.00 US plus shipping
Ordering instructions below
Description: The Bird Model # 43 Thruline Wattmeter is a very common test tool among Repeater Builders worldwide. The dummy slug can be affordably machined into a very useful tool for testing desense, monitoring waveforms, and feeding other equipment. This product does not disturb the original intent of the dummy slug; keeping out the dust and shunting the meter movement in travel. It allows coupling into the system without needing to install an Iso Tee or other such tool which can change line lengths or impedance, allowing you to optimise your set-up and knowing the achievement will stay after the device is removed from the circuit.
This unit is machined with a lathe and end mill to produce a professional job. The sampler uses a precision 51 Ohm resistor connected to a BNC connector to provide the coupling. Its usable range is 50 MHz. to over 2.5 GHz.
is a PDF document which contains VNA graphs showing the coupling losses
of the sampler when rotated for maximum and minimum coupling.
NOTES: The following notes have been offered by Allan Crites - WA9ZZU
When using an iso-tee in this instant application it is highly desireable to use a 50 ohm termination at the iso-tee to properly terminate the signal source. Using a modified & terminated Bird Wattmeter slug for signal sampling is a highly desireable means for observing off the air signals, for observing desense, and transmitter output. However, there is the possibility of the transmitter output entering the signal generator and creating intermod if used for the purposes of making a desense measurement and should be seriously considered. I recommend the use of a 10 dB attenuator on the BNC output to the signal generator to lessen the chance of intermod. While rotating the slug reduces any transmitter signal it also reduces the coupling that would be required for the signal generator to overcome. Another way to reduce the coupling would be to not insert the slug as far into the Wattmeter, thus having the same effect/benefit as the iso-tee isolation with the added characteristic of a 50 ohm termination on the BNC connector. This tool has the added benefit of directional magnetic coupling whereas the iso-tee is capacitive and does not. AC
Contact Kevin for availability
- $35.00 US plus shipping - PayPal preferred.
to the top of the page
Up one level
Back to Home
and HTML'ized by Kevin K. Custer W3KKC
Artistic layout and hand-coded HTML © Copyright August 26, 2011 and date of last update by Kevin K. Custer W3KKC
This page last updated January 13, 2012 - W3KKC
web page, this web site, the information presented in and on its pages
and in these modifications and conversions is © Copyrighted 1995 and
(date of last update) by Kevin Custer W3KKC and multiple originating authors.
All Rights Reserved, including that of paper and web publication elsewhere.